Gammino S., Celona L., Ciavola G., Castro M., Chines F., Marletta S.
INFN-Laboratori Nazionali del Sud, Via S. Sofia 44, 95123 Catania, Italy
Bieth C., Sortais P., Kantas S.
Pantechnik S.A., rue A. Kastler 12, 14000 Caen, France
The source ECR2 has been built in 1998 by Pantechnik, according to the design suggested by LNS Ion Source Group. This design entails some improvements with respect to a standard CAPRICE-type source:
a) the magnetic field (up to 1.6 T axial, 1.1 T radial) allows to operate the source at 14 GHz in High B mode and at 18 GHz;
b) two frequency heating can be used;
c) an aluminum made plasma chamber is used in place of the stainless steel one.
The main features of ECR2 along with a review of the preliminary tests will be outlined. Typical currents for fully stripped nitrogen are about 25 emA; for the heaviest ions, 1 emA of Kr28+ and 10 emA of Ta27+ have been measured.The installation at LNS has been completed recently and the details will be given.
Presenting Author : Gammino S.
Presentation : Oral